As a new type of medical equipment, the stretcher cart is different from the general hospital movable bed in terms of convenience. In order to alleviate the pain index of patients and the convenience index of medical care, the index has been greatly improved. However, there is not only one type of stretcher. It can be divided into several types according to different functions and places of use. The following are some of the more common types of stretchers.
The first type of medical stretcher cart belongs to the technical field of medical equipment. The current stretcher trolley has a single function, and there is no infusion rack or drainage net, so it brings a lot of trouble to the medical staff. There must be several people holding the infusion rack, drainage net, and clothes and quilt to follow the stretcher trolley before it moves, thus it's really inconvenient. Compared with the previous one, this Ambulance Stretcher greatly brings convenience to medical staff and relieves the pain of patients.
The advantages of the utility model are: the structural design is reasonable and practical, because the infusion rack, the wardrobe, and the drainage net are arranged on the frame to reduce the number of medical personnel accompanying the stretcher cart, reduce the labor intensity, and bring a lot of conveniences. This type of stretcher is specially designed for hospitals.
The second is a utility model of a multi-purpose stretcher cart, which consists of an upper frame, a wheel bracket, an axle mounted on the wheel bracket, the axle is equipped with wheels and a rear bracket. This multi-purpose stretcher is more suitable for ambulances, aka, ambulance stretcher, it’s light, foldable, and easy to carry.
It can realize the functions of a stretcher cart and an individual soldier trailer by changing the different folding methods of the wheel bracket, auxiliary bracket, and rear bracket that are hinged on the upper frame. It is also very convenient to carry and use.
The third one is a new type of convenient stretcher trolley, which can be used with all docking trolleys. There is no need to worry about the cart model and the stretcher trolley. It is only necessary to set a concave rail groove in the middle of the borders of the two sides of the stretcher bed for the docking trolley and the bottom of the groove. There are magnets, which match the two magnetic pulleys on the back of the stretcher, that is, they are all inserted from the left and right sides of the wounded, and the bandages are fastened at both ends. It is easy to operate, saves manpower, and facilitates promotion in hospitals.
The fourth type is an ambulance stretcher cart with automatic damping and buffering up and downstairs, especially when patients live in high-rise residential areas. It consists of a bracket and a rotating shaft, a damping buffer, and a damping buffer frameset at the bottom of the middle of the stretcher. One end of the stretcher is provided with a handle, a support frame, and an auxiliary wheel, the other end of the stretcher is provided with an unlocking handle, a connecting rod, and a locker, and there is a stretcher cloth on the stretcher. Up and downstairs are controlled by the unlocking handle to rotate the damping buffer wheel and the driving wheel, and the damping buffer wheel and the damping driving wheel are folded through the rotating shaft. You can walk up and down the stairs easily.
Its advantages are that the structure is simple and the operation is convenient, it’s not bumpy when you go up and downstairs and thus very stable, the whole stretcher body is flexible when it turns in all directions, which saves time and effort and can be operated in time and rescue patients in a short time. This type of equipment is often widely adopted in hospitals when patients go up and downstairs, as well as on and off the ambulances. The stretcher-type patient carrier can be panned left and right and folded. This kind of stretcher belongs to the technical field of medical equipment. There is a guide rail at each end of the upper part of the car body, and there are guide sleeves on the guide rails, which can slide along the guide rails, and can roll the sheets under the patient's body into a stretcher. The stretcher can stand vertically to one side when not in use. The wheels of the car can be retracted, thus reducing the floor space.
The above are the various types of stretcher carts and their performance characteristics. As a medical device, stretcher trolleys play a very important role, so medical staff must be careful in their choice, a regular and reliable stretcher trolley manufacturer is unavoidable. Buying genuine products to ensure quality and quantity. Stretchers are critical for patients, it will cause secondary injury if a stretcher is festered with design defects or quality problems.
“Basket stretcher” is a term used by emergency responders, especially in the context of a medical emergency. It refers to a device that is designed to support and lift a victim’s head and neck out of a basket (such as in an ambulance) and off of the floor. The basket stretcher is used to help remove injured people from their vehicles, or to transfer them from an ambulance to another vehicle, like a private vehicle.
Basket stretchers are not completely known for their safety. Although they are safe when used properly without causing injury, there have been reports of them injuring rescuers when they fall on them. Their use can also lead to accidental injuries for rescuers who grab the stretcher too tightly or are not careful enough with it; this could be especially dangerous if someone gets pinned underneath it. An important consideration when using basket stretchers is that it should be positioned away from moving parts (such as the engine). If it is in contact with the engine, it could cause engine burnout problems if the engine starts unexpectedly (and if an accident occurs while this happens, rescuers could end up getting burned).
Basket stretchers should never be used until after it has been thoroughly tested by professionals and evaluated so that any potential issues can be addressed before use. This way, you will benefit from knowing what can go wrong before you make your mistake in using these devices; this also helps you avoid making similar mistakes again at another time.
This is a topic that is often neglected and so goes largely unmentioned. Given the extreme importance of the basket stretcher, it is odd that this has been ignored for so long. Let me explain:
Basket stretchers are typically made of sturdy materials such as aluminum or steel (hence the name). They are, in general, very rugged and well-made. They may also be made of wood, plastic or paper to facilitate them being carried and/or stored in a small space. This presents a major problem:
They are not collapsible.
This means that if you have to go away for an extended period of time (e.g., to the hospital) , you may not have access to your basket stretcher at all during that extended time – even if you can’t carry it with you (the size and weight will be too much for your carry bag). You need to bring it with you on the way out and take it with you on the way back in – which requires additional lifting action and effort in addition to your normal lifting actions during each trip home / work / etc. It’s absolutely crucial not to do something stupid while using such a tool – I know I’ve been there! We have used basket stretchers as part of our daily life for well over 20 years now, but we don’t do anything special with them (other than put them away when we are finished using them). We use them just like any other tool: lift one end into place, load up any stuff we need with our baskets on top, let go, walk away!
On top of all this hassle – because baskets are made of lightweight material – they can be easily punctured by heavy objects; e.g., sharp objects like knives or forks which would damage their material as well as any other part of the tool; e.g., blades which can cut through their material if they get jammed inside their baskets while they are being used; or even fists if they get jammed between the handles while in use (which again could damage both their material and your hands); or a combination of both those things:
They can puncture their baskets under use due to pressure applied by fingers squeezing inside the handles when they lift something into place; e.g., when picking up an empty coffee cup from your cupboard; or just by getting zippered closure items stuck outside their basket while being carried/being pulled along by.
This post is a hypothetical question from our community, with the intent of illustrating the potential risk of using a basket stretcher .
The basket stretcher is a very useful tool for stretchers. It helps to reduce the risk of the patient being transported to the operating room in an upright position and to reduce the risk of injury in the operating room. However, if used incorrectly, it may increase these risks. Because it is used with care, some caregivers have minimized its use in order to avoid any risk.
A basket stretcher is a device that is used to transfer a basket from one area to another. It is a very simple, cheap and easy tool to use and it can be used in a number of ways. In brief, what you need to do with basket stretcher:
Set it on the floor and choose an appropriate place for the basket
Push the top of the basket stretcher against the base of your furniture, or your legs – whichever fits better for you
Place the undergarments or other support items in the basket, or place them on top of it
Use your hand or something else to spread the undergarments so that they fit evenly
Take them off and let them dry
Replace them and repeat until you are satisfied with how it looks
Store it somewhere where moisture can’t get inside
• Store at room temperature – this will keep it in good shape over time
• It should be kept clean – if dirt gets inside it, mold will grow inside and cause damage
• It should be used only outdoors – if left indoors, mold can form there too
• The baskets should not be made out of polyester (or nylon), which will rip when they are stretched
Firm pressure on the head position control handle releases the self-locking gas spring, extending the piston rod and simultaneously slowly raising the bed surface from the head position. Also hold the handle tightly and apply downward force to descend;
The lifting and lowering of the thigh bed surface is controlled by the thigh crank handle; the lifting and lowering of the foot bed surface is controlled by the foot position control handle.
When gripped tightly, the pulling pin is released from the positioning hole and lowered by its own weight. When the handle is released, the foot bed surface 12 is locked in this position when it reaches the desired angle;
The coordinated use of the control handle and the handle can enable the patient to obtain various postures from supine to semi-recumbent, bent legs, flat sitting, and upright sitting.
In addition, if the patient wants to lie on their side in the supine position, first pull one side of the hospital bed from supplier, lower the side rails, and press the control button on the outside of the bed with one hand to release the self-locking side gas spring, its piston rod extends.
At the same time, the lying surface on the side is slowly raised. When it rises to the required angle, release the control button to lock the bed surface in this position, and complete the side lying position from the surface. Also the same operation is used in reverse.
When the defecation locker is rotated clockwise, the toilet hole cover will automatically open, and the toilet carrier will automatically feed the patient's buttocks horizontally, allowing the patient to defecate or clean the lower part. keep it flat. Keep it flush.
At the same time, the bedpan is automatically sent to the operator's side, so that the nursing staff can take it away for cleaning, and put the cleaned bedpan back into the bedpan carrier for the next use.
Hold the top edge of the side guardrail horizontally, lift it up about 20 mm vertically, and turn it down 180 degrees to lower the guardrail. After patient goes to hospital bed, After the patient goes to bed, lift the guardrail up and turn it 180 degrees, and press it vertically to complete the side guardrail raising. Note: The same way as foot guards used.
Align the opening of the plastic part on the back of the living table with the upper side of the side rail and press it down. Hold the handrail with one hand and lift the living table with the other to remove it.
No matter the state of the bed surface, the infusion pole can be used. When using the infusion wand, first twist the two parts of the infusion wand into one.
Then bend the hook under the infusion stand, align it with the upper horizontal tube, and at the same time, align the upper hook head with the round hole of the upper tube of the side guardrail and press down to use it. Lift the infusion stand and you can remove it.
Press the brake with your feet or hands to brake. Lift and remove the infusion handle.
When the patient uses the bed or needs to change the body position, he must fasten the seat belt (the tightness of the seat belt is adjusted according to the individual) to prevent danger.
When the foot bed surface is level, adjust the thigh handle and lift the thigh bed surface to prevent the patient from slipping; hold the foot control handle tightly, put the foot bed surface in the proper position, and turn down the foot activity board. Then shake the thigh handle to keep the foot activity board level, put it in the water basin, and you can wash your feet. After washing your feet, remove the basin and replace the foot pads. Hold the foot control handle and raise the soleplate to a horizontal position.
In the simplest case, China hospital beds from manufacturer can be used to sit on. Lower the bed to a position where you can rest your feet comfortably on the floor.
You can sit on the bed to have a rest from standing, or for intermittent fetal monitoring. Sitting in bed is much more comfortable than going into labor on your back. It can also be used for epidural sitting in bed. Off course upu can sit on the bed, with your legs and feet out flat or sitting cross-legged.
How to choose an electric wheelchair? The first thing we need to consider is that electric wheelchairs are for users, and each user's situation is different. From the perspective of the user, based on basic data such as height and weight, daily needs, use environment and special surrounding factors, etc., a comprehensive and detailed assessment should be carried out from the perspective of the user, so that effective selection can be carried out, gradually subtracted until a suitable electric wheelchair is selected.
How to choose an electric wheelchair? In fact, the conditions for selecting electric wheelchairs and ordinary wheelchairs are basically similar. When selecting the seat back height and seat width, the user can sit on the electric wheelchair, the knees are not curved, and the calf can be naturally lowered at a 90° right angle. The appropriate width of the seat is, at the widest position of the hips, add 1-2cm to the left and right.
If the user sits, the knees are slightly high, and the legs will be curled, and it is very uncomfortable to sit for a long time. If the seat surface is narrow, the sitting will be crowded and wide, and long-term sedentary will cause secondary injuries such as spinal deformation.
Then consider the weight of the user. If the weight is low, the use environment is flat, and the brushless motor is cost-effective; if the weight is heavy, the road condition is not very good and needs to drive long distances, it is recommended to choose a worm gear motor (brushed motor).
The easiest way to detect the power of a motor is a climbing test, whether the climbing test motor is easy or slightly strenuous. Try not to choose the motor of the horse-drawn cart. There will be many faults in the later stages. If the user has more mountain roads, it is recommended to use a worm motor.
The battery life of an electric wheelchair is also a concern for many users. To understand the properties of the battery and the AH capacity, if the product description is about 25 kilometers, it is recommended that the budget life be 20 kilometers, because the test environment and the actual use environment will differ. For example, the battery life in the north will be reduced in winter, and try not to drive an electric wheelchair out during colder times, which will cause great and irreversible damage to the battery.
6AH battery life 8-10 km
12AH battery life 15-20 km
20AH battery life 30-35 km
40AH battery life 60-70 km
Battery life is related to battery quality, weight of electric wheelchairs, passenger weight, and road conditions.
How to choose an electric wheelchair? It is also important to note the overall width of the electric wheelchair. Some families have narrow doors and need to measure the width and choose an electric wheelchair that can be freely accessed. Most electric wheelchairs have a width of 55-63cm, and some have made it more than 63cm.
When buying products, you need to carefully understand the instructions, check whether the product label brand is consistent with the manufacturer. If the product label brand is not consistent with the manufacturer, it is an OEM product.
Finally, let's talk about the warranty time. Most of them are one-year warranty for the whole vehicle, and there are also separate warranties. The controller is regular for one year, the motor is for one year, and the battery is 6-12 months.
There are also merchants who have long warranties, and ultimately follow the warranty instructions in the manual. It's worth noting that some brands have warranties based on the factory date, and some are based on the sell-by date.
When buying, try to choose a production date that is closer to the date of purchase, because most electric wheelchairs have batteries that are installed directly on the electric wheelchair and are sealed and stored and cannot be maintained separately. If left for a long time, the battery life will be affected.
Friends who have been using electric wheelchairs for a long time may find that the battery life is slowly short, and an inspection finds that the battery drum is packed. Either you run full and run out of power, or you can't charge it how you charge it. Don't be in a hurry, tell you how to properly maintain your battery today.
In the driving state of the electric wheelchair, the battery itself will heat up, plus the weather is hot, the temperature of the battery is even as high as 70 ℃. In the case of the battery not cooled to the ambient temperature, the electric wheelchair will be charged immediately as soon as it stops, which will exacerbate the battery's lack of water, reduce the service life of the battery, and increase the risk of battery charging.
It is recommended to park the electric vehicle for more than half an hour and charge it after the battery is fully cooled. If the battery or motor is abnormally heated during the driving of the electric wheelchair, please go to the professional electric wheelchair maintenance department for inspection and repair in time.
The battery also heats up during charging, and if charged under direct sunlight, it can also cause the battery to lose water and cause a drum bag to the battery. Try to charge the battery in a cool place or choose to charge an electric wheelchair in the evening.
Charging an electric wheelchair with a mismatched charger may result in damage to the charger or damage to the battery. For example, using a charger with a large output current to charge a small battery can easily cause the battery to charge the drum pack.
It is recommended to go to a professional electric wheelchair after-sales repair shop to replace the matching high-quality brand charger to ensure the charging quality and prolong the service life of the battery.
Many electric wheelchair users often charge all night for convenience. The charging time often exceeds 12 hours, and sometimes forgets to cut off the power to charge for more than 20 hours, which will inevitably cause great damage to the battery. Charging multiple times for a long time can easily cause the battery to charge the drum due to overcharging. Generally, electric wheelchairs can be charged for 8 hours with a matching charger.
Try to keep the electric wheelchair battery fully charged before traveling, and depending on the actual range of the electric wheelchair, you can choose to take public transportation for long-distance travel.
After reading this article, do you have a general understanding of how to choose an electric wheelchair?
Subscribe to our email list and stay upto date with all your latest updates